Calibration plate(s) included in set can be also used to check up correctness of measurement
After applied device's probe to the calibration plate with "FE-200um" / "AL-200um" description paint meter should shows 200 µm result on the display (+/- 10 µm measurement error).
Calibration Plate Steel FE-200µm
Calibration Plate Aluminum AL-200µm
Notice! Paint meters without possibility of measurement aluminium has only one calibration plate with "FE-200um" description in set.
Calibration plate "AL-200um" is only attache to paint meters with the measurement of aluminum function. Coating thickness gauge with measurement of aluminum function: MGR-10-S-AL, P-10-S-AL, MGR-11-S-AL, P-11-S-AL.
Calibration procedure for series: A-10-FE
In the case of incorrectly measurement results paint metershould be calibrate. Please use attached calibration plate to the calibration process.
When the device is turn off press and hold button through 10 seconds. The display should show the numbers sequence from 90 to 10.
If you see on the display "HL" and "FE" description please relase button.
Please put and hold probe to the grey calibration plate and press buton.
NOTICE! When you press button during calibration procedure, probe must be adjoin to the calibration plate!